High Performance Liquid Chromatography in analysis
MS Ramaiah college of Pharmacy, Rajiv Gandhi university, Bengaluru, Karnataka, India
- *Corresponding Author:
- T.N. Divya, MS Ramaiah college of Pharmacy, Rajiv Gandhi university, Karnataka, India
E-mail: [email protected]
Received date: 15/02/2014 Accepted date: 16/03/2014
Visit for more related articles at Research & Reviews in Pharmacy and Pharmaceutical Sciences
Chromatography, HPLC, Analysis, Column
High Performance Liquid Chromatography, also known as high pressure liquid chromatography is the most commonly used analytical methods in the pharmaceutical industry. High Performance Liquid Chromatography (HPLC) is mainly used for the separation, identification and qualitative & quantitative analysis of the components in a sample mixture. Usually it is used in the case of semi-volatile or nonvolatile organic compounds [1,2].
It is a type of column chromatography but is highly sensitive and extremely automated compared to the column chromatography. Unlike the solvent allowed under gravity through a column in column chromatography, the HPLC technique utilizes high pressure up to 400 atm [3-5]. It allows faster and better separation than the column chromatography due to the pressure and the greater surface area of the packing material in the column which is due to the smaller particle size accommodated. This greater surface area facilitates better interactions between the packing material which is the stationary phase and the molecules to be separated which flow through the stationary phase [6-9].
Operation of the HPLC technique is a little confusing. It involves a sampler, pump and a detector. HPLC uses the concept of affinity between the mobile phase which is the solvent and stationary phase [10,11]. A pump is used to inject a continuous flow of solvent. Into this solvent a dissolved sample is injected. Once the sample is in the solvent, it passes through the column [12-14]. The components in the sample are separated based on their affinity to the molecules in the column. After the compounds in the sample are separated, they pass the detectors. User software is utilized in the HPLC technique for the data analysis [15-18].
The HPLC technique can be used in two forms ;
Normal phase HPLC
This is the basic column chromatography. It is less frequently used form of HPLC
The column is filled with silica molecules, and the solvent used is non-polar.
Polar components in the mixture being passed through the column will adhere longer to the polar silica particles than non-polar compounds. The non-polar compounds will therefore pass more quickly through the column [20,21].
Reversed phase HPLC
Reversed phase HPLC is the most frequently used form of HPLC.
In this case, the silica is made non-polar by attaching long hydrocarbon chains to its surface. A polar mobile phase is used .
Therefore, there will be a strong adherence between the polar solvent and polar molecules in the mixture being passed than between the hydrocarbon chains attached to the silica and the polar molecules in the solution .
Non-polar compounds will form attractions with the hydrocarbon groups because of van der Waals dispersion forces. They are less soluble in the solvent. Therefore this will slow them down on their way through the column. Now, the polar molecules will pass through the column more quickly [24,25].
- Tyagi A, Sharma N, Mittal K, Mashru R, Bhardwaj T, et al. (2015) HPTLC-Densitometric and RP-HPLC Method Development and Validation for Determination of Salbutamol Sulphate, Bromhexine Hydrochloride and Etofylline in Tablet Dosage Forms. Pharm Anal Acta 6:350.
- Lu Y, Tian L, He Y, Lu Y, Liang X, et al (2015) Development and Optimization of a RP-HPLC Method to Quantify Midazolam in Rat Plasma after Transdermal Administration: Validation and Application in Pharmacokinetic Study. Pharm Anal Acta 6:329.
- Nardulli P, Capparelli E, Perrone MG, Ferraiuolo S, Laforgia MR (2015) A Combined HPLC and LC-MS Approach for Evaluating Drug Stability in Elastomeric Devices: A Challenge for the Sustainability in Pharmacoeconomics. J Pharmacovigilance 3:157.
- Singh A, Tandon S, Sand NK. (2014) Active Ingredient Estimation of Clopyralid Formulation by Reversed Phase HPLC. J Chromatogr Sep Tech 6:257.
- Myron P, Siddiquee S, Azad SA, Yong YS. (2015) Tributylamine Facilitated Separations of Fucosylated Chondroitin Sulfate (Fucs) by High Performance Liquid Chromatography (HPLC) into its Component Using 1-Phenyl- 3-Methyl-5-Pyrazolone (PMP) Derivatization. J Chromatogr Sep Tech 6:256.
- Sassi A, Hassairi A, Kallel M, Jaidane M, Saguem S. (2014) HPLC Method for Quantification of Halofuginone in Human Ureter: Ex-Vivo Application. J Chromatogr Sep Tech 6:255.
- M Sangeetha, C RubinaReichal, VN Indulatha, N Thirumoorthy (2014) Development and Validation of RP-HPLC Method: An Overview. Research and Reviews: Journal of Pharmaceutical Analysis 3.
- Javed Ahmad, KanchanKohli, Showkat R Mir, Saima Amin (2013) Development and Validation of RP-HPLC Method for Analysis of Novel Self-emulsifying Paclitaxel Formulation. Research and Reviews: Journal of Pharmaceutical Analysis 2.
- Lovekesh Mehta, Jitender Singh (2013) RP-HPLC Method Development and Validation for the Determination of Bupropion Hydrochloride in a Solid Dosage Form. Research and Reviews: Journal of Pharmaceutical Analysis 2.
- Ezhilarasi K, Sudha V, Ramachandran G, Umapathy D, Rajaram R, et al. (2014) A Simple and Specific Method for Estimation of Lipoic Acid in Human Plasma by High Performance Liquid Chromatography. J Chromatogr Sep Tech 5:245.
- Shintani H (2014) Role of Metastable and Spore Hydration to Sterilize Spores by Nitrogen Gas Plasma Exposure and DPA Analysis by HPLC and UV. PharmaceutReg Affairs 3:125.
- Malferrari M, Francia F (2014) Isolation of Plastoquinone from Spinach by HPLC. J Chromatogr Sep Tech 5:242.
- Naveed S (2014) Analytical Determination of Lisinopril Using UV Spectrophotometer and HPLC: An Overview. Mod Chemappl 2:137.
- Shintani H (2014) Serum or Saliva Extraction of Toxic Compounds from Methyl Methacrylate Dental Materials and HPLC Analysis Combined with SPE. PharmaceutReg Affairs 3:123.
- Rudraraju AV, Hossain MF, Shrestha A, Amoyaw PNA, Tekwani BL, et al. (2014) In vitro Metabolic Stability Study of New Cyclen Based Antimalarial Drug Leads Using RP-HPLC and LC-MS/MS. Mod Chemappl 2:129.
- Tang M, Nguyen J, Sadoun M, Dorin Ruse N (2014) HPLC Analysis of Monomer Release from Conventionally and High Temperature High-Pressure Polymerised Urethane Dimethacrylate Intended for Biomedical Applications. J Chromatograph SeparatTechniq 5:227.
- Elshanawane AA, Abdelaziz LM, Mohram MS, Hafez HM (2014) Development and Validation of HPLC Method for Simultaneous Estimation of Brimonidine Tartrate and Timolol Maleate in Bulk and Pharmaceutical Dosage Form. J Chromatograph SeparatTechniq 5:230.
- Mustafa S, Hashim W, Khaliq S, Azizuddin, Khan RA (2014) An Improved High Performance Liquid Chromatographic Method for Tryptophan Analysis in Rat Brain Administrated by Seaweed. J Anal Bioanal Tech 5:188.
- Caglar S, Alp AR (2014) A Validated High Performance Liquid Chromatography Method for the Determination of Saxagliptin and Metformin in Bulk, a Stability Indicating Study. J Anal Bioanal Tech S12:010
- Abdallah MA (2014) Validated Stability-indicating HPLC and Thin Layer Densitometric Methods for the Determination of Pazufloxacin: Application to Pharmaceutical Formulation and Degradation Kinetics. J Chromatograph SeparatTechniq 5:218.
- deFigueiredo NB, Oiye ÉN, de Menezes MMT, de Andrade JF, Brunini Silva MC, et al. (2010) Determination of 3,4-methylenedioxymethamphetamine (MDMA) in Confiscated Tablets by High-Performance Liquid Chromatography (HPLC) with Diode Array Detector. J Forensic Res 1:106.
- Shah I, Barker J, Barton SJ, Naughton DP (2014) A Novel Method for Determination of Fenofibric Acid in Human Plasma using HPLC-UV: Application to a Pharmacokinetic Study of New Formulations. J Anal Bioanal Tech S12:009.
- Gurupadayya BM, Disha NS (2013) Stability Indicating HPLC Method for the Simultaneous Determination of Ceftriaxone and Vancomycin in Pharmaceutical Formulation. J Chromatograph SeparatTechniq 4:207.
- Paranthaman R, Kumaravel S (2013) A Reversed-Phase High- Performance Liquid Chromatography (RP-HPLC) Determination of Pesticide Residues in Tender Coconut Water (elaneer/nariyalpani). J Chromatograph SeparatTechniq 4:208.
- Shintani H (2013) HPLC Separation of Amino Acids is Appropriate? Pharmaceut Anal Acta. 4:e158.